Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||14 December 2010|
|PDF File Size:||6.74 Mb|
|ePub File Size:||19.27 Mb|
|Price:||Free* [*Free Regsitration Required]|
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is mwnual set at.
If not, what would creating a driver for it entail? Message 1 of 6. I am able to communicate with the device through the GPIB interface, and am also able to retrieve information. When I gave it a try, though, it worked fine.
I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different. It’s 84MB, so I hope you have a fast link. Message 3 of 6. I have worked with ho4191a A and, as I recall, the commands are nothing like any other instrument I have seen.
HP A for Sale|Impedance Analyzers|Impedance Measurement Products|Test Equipment Center
Has anyone done anything on interfacing with the A since Matt C’s last post? Message 4 of 6.
You really need the manual. Would you mind posting the code you used on the A?
The forums emoticons took over for a bit there. Matt – you’re right, the commands for the A are hp419a different from the – probably because of its age.
Agilent / HP 4191A RF Impedance Analyzer
Message 6 of 6. Sorry, that last line of code there is supposed to read: Message 5 of 6. I suspect that this difference might be causing my problems.
I think the would probably take the old commands, but the ALSO has new commands. Guess you never know until you try! Most Active Software Boards: Good luck with it Should I be looking at doing it with LabView or something else?
IIRC, the sweep parameters were something like this: In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:. Thanks for your help guys. In particular, after setting hp41991a and initiating the instrument, I am initiating the measurement and reading it as follows: Accepted by topic author Matt C. In the code above, MHz is an arbitrary frequency.
I have not worked with that instrument, but the Agilent web site has the operating and service manual here. Message 2 of 6.