Model: A. Manufacturer: Fluke. About: We are NES Sales, Northeast’s industrial source. Notes: Does not include stand. Sold as-is, as pictured. | eBay!. Fluke A bench multimeter. «on: November 04, , am». I got this oldie from ebay the other day, the seller sold it as untested but it works. Fluke will not be liable for any claims, losses or damages of any kind incurred by any user arising from use of this manual. A Digital Multimeter Instruction.
|Published (Last):||5 March 2018|
|PDF File Size:||8.43 Mb|
|ePub File Size:||4.39 Mb|
|Price:||Free* [*Free Regsitration Required]|
Select the DCV function and the 2V range. If the recommended equip- ment is not available, equipment of equivalent specifi- cations may be used.
Fluke 8600A Portable Bench Digital Multimeter 1000v DC
Shipping cost cannot be calculated. The Battery Pack provides the A with the capability of operating as a portable battery-operated in- strument.
If absent or wrong, check that output from U8, keeping in mind that the driver transistors act as inverters. Auto- ranging can be selected when the ac volts, dc volts or kil- ohms functions are selected.
Minimum monthly payments are required. A Gulf Western Mfg.
Digital Multimeter 8600A
The A can be operated from1 15, or V ac 50 or 60 Hz power lines, as selected by line power select switch S The following information is provided to assist in locating malfunctions in the A. A dc offset voltage created within the ac converter may cause the multimeter display to indicate some substantial value of ac voltage when a short is placed across the input terminals.
Four strobe cycles are required to load the new data. If the waveform is correct, the problem is in the display section. F,uke More – opens in a new window or tab Any international shipping is paid in part to Pitney Bowes Inc. Sub of Em hart Industries Inc.
Replace the batteries with 1. BZ from the instrument to the DOU occurs synchro- nously with the second STO signal affer a measurement is complete and lasts for one strobe cycle. To ensure prompt and efficient handling of your order, include the following information. At the end of the integrate period Q14 is turned off, U4 and U5 no longer charge C28, and the charge on C28 is held.
Refer to Section 6 Option —01 for requirements of line voltage change for A instruments. The flluke and function switch posi- tions in conjunction with flu,e signal inputs to U16, con- trol the illumination of the polarity indicator on the display and the selection of five or six ranges for the autorange feature.
Fluke A Digital Multimeter | eBay
Sell now – Have one to sell? Ten volts is maintained across the input divider in all ranges except the 20 range. If you need any further information about this item, please ask any questions prior to purchase. To gain access, remove ac power, remove the retaining screw on the rear of the case and remove the instrument from the case. Push-button controls allow the selection of five ac and dc voltage ranges, five ac and dc current ranges, and six resistance ranges.
Connect the high input lead to TP4, the volt supply. Observe the voltage reading displayed in volts rms on the A readout.
Fluke DMM A repair photos – Mobile Modding
Contact the seller – opens in a flukd window or tab and request a shipping method to your location. Sub of Raytheon Co. The A-D Converter uses a dual-slope conversion technique. High Voltage Probe, Schematic Clean the front fouke and exterior surfaces with an- hydrous ethyl alcohol or a soft cloth dampened with a mild solution of detergent and water. The front panel dust cover is a molded plastic snap-on accessory which fits over the front panel of the A.
Electrical components are listed by reference designation and mechanical components are listed by item number. The rear panel power input connector is a three- prong, U-ground connector which permits the instrument to be connected, via the 6800a cord, to the appropriate line power.
Input Signal Conditioners Framingham, MA Teledyne Inc. Fluke will not be liable for any claims, losses or damages of any kind incurred by any user arising from use of this manual.
These voltage limits may be used in combination so that the ac component of an ac signal superimposed on dc level can be measured. Wesson, MS Thermalloy Co.